The PlasmaTherm SLR has 2 ICP processing chambers plus a transfer chamber and a load-lock.
The tool will normally be set up for 100 mm (4") wafers. However, it is quite easy to change the wafer clamping to 150 mm (6").
It is not allowed to etch two wafers glued together! It will not be flat enough and will cause the clamp to break.
It is allowed to glue smaller pieces onto a carrier wafer if the pieces are inside the clamping ring and stay there.
The right chamber is used for Deep Reactive Ion Etching (DRIE). It is primarily intended for etching deep features in silicon by means of the Bosch process.
The etch rate will depend strongly on the feature size and the amount of exposed area. A typical value is an etch rate of around 3 µm/min.
Typical selectivities for silicon etching are 80 for photo resist and 200 for silicon oxide. An aluminum mask has a very high selectivity and may be used for etching through a wafer.
The left chamber is intended for metal etching using chlorine chemistry. It is mainly used for aluminum etching.
Other metals and metal compounds may be etched in the left chamber after discussions with MSL.